Mini-spectrograph was designed to depict the spectrum of received light on the sensor of internal camera .
The constructions details:
The opening for diaphragm: 32 mm width, 1 mm thick. The size of the hole in the diaphragm should be a tradeoff between the resolution of the plot and the intensity of measured light
the distance between slit diaphragm and lens = 100 mm
the collimator lens: F = 100 mm, D = 25 mm
the optical gratting: 1000 lines pro mm
The handle is special adapted to Nexus-5. We are preparing the new design of spectrograph, splitted into 2 components: spectrograph and montage-adapter. The adapter will be prepared for the most popular Android devices
3D-Print: Spectroscope is published in MAKE as DIY project
Optical grating: Rainbowsymphony diffraction slide
Colimator lens: Thorlabs LA1251
Slit diaphragm: Cardboard, or plastic with the small hole.
Dietrich Hubert who helped me to construct the spectroscope
Matthias Kunkel, Philipp Nimtsch, for inspiration and help in the optics